# A Useful Current Profiling Method

In one of my recent projects, I had to capture the dynamic current profile of a short-range wireless embedded platform for a number of operation scenarios. Due to the dynamic nature of the current consumption involved, I was unable to use a standard voltmeter for this job. On the time axis, I needed a minimum resolution of 1 millisecond and the accuracy of the current measurement was supposed to be better than 2 mA. This task would be a piece of cake with a Digital Storage Oscilloscope, which I did not have at the time and therefore I had to come up with a cheaper method to get the job done.

The top-level diagram of the solution I came up with is shown below. I used 3.0 V button cell battery as my power source in all of my measurements, however you can use an ordinary DC power supply in your measurements as required by the particular load you use. The load is the device whose current profile we are interested in. The load’s positive terminal is connected to the battery and the ground terminal is in series with a 0.3 ohm shunt resistor (Rs) that is terminated at ground. The shunt resistor needs to have a small value in order not to cause a large voltage drop across the Rs, which decreases the required operational voltage across the load (You can think of Rs as an ordinary wire with a bit of resistance). The power of Rs is usually a high value (e.g. 1 Watt) in order not to get damaged by the current flowing through it. The voltage across the shunt resistor (denoted Vs) varies depending on the current drawn by the load. This current is simply equal to Vs/0.3 amperes.

Note that Vs is typically small due to the small value of the shunt resistor. Therefore, digitisation and the measurement of Vs directly is difficult and an intermediate amplification stage is necessary. In my application, an amplifier with a gain of ~11 was used. If you need higher gains in your measurements, simply adjust the R2/R1 ratio as needed. The overall gain for the amplifier topology provided is ( 1 + [R2/R1] ). I simply used a general-purpose LM324 opamp which worked really well.

The output of the amplifier is fed to an analog-to-digital converter (ADC) that had a resolution of 10 bits per sample and ran at a sampling frequency of 20 kHz. This sampling rate equates to 50 microseconds between successive samples, which is better than what I needed. Higher sampling rates can provide much better resolution in time but remember that your storage requirements will also increase as a result. The digital output of the ADC is converted to milliamps and for each sample a time stamp in milliseconds is also associated. Note that the blocks indicated inside the dotted lines are an integral part of the microcontroller platform (i.e. AtMega2560) I used in my application.

In my application, I dumped the SRAM content after each measurement session due to the memory limitations of the platform I used. However, I also experimented with real-time serial dump of each ADC sample and that method will also work if your measurement resolution requirements are more relaxed. You can easily use a serial terminal program such as TeraTerm to save the ADC samples.

Using the above set up, it is possible to sample current levels as low as 1.15 mA. Figure below illustrates a captured current profile during the RF transmission of the load I used in my project. The repetitive transmission tasks can easily be identified using the dynamic current profile as shown in the figure.

I hope you will find my tried and tested method for current measurement useful in your projects. The best part of it is it allows you to get accurate measurements without an expensive digital storage oscilloscope!

[ - ]
Comment by July 2, 2012
A good overview of testing of this type. I'd add that Rs for current levels of this type usually isn't chosen as a high-wattage resistor, but rather on the basis of voltage drop that takes away from the battery voltage. Even at 100mA, a 0.3 ohm resistor has an IR drop of 30mV and an I2R power dissipation of 3mW. Nowadays even the 0603 resistors are rated at 100mW power dissipation, so neither the IR drop nor power dissipation should be a problem here. In power supply circuits when you get to hundreds of milliamps, then you have to worry more about sense resistor wattage.
[ - ]
Comment by July 3, 2012
How accurate do you need the current profile? After a cascade of low pass filters, like the Rs and the Opamp, it would be only natural to have your high freq components gone. Are these accounted for?
[ - ]
Comment by July 3, 2012
I was not interested in the current profile changes above 10 kHz for this application as obtaining an average current consumption was more important than capturing the peak and the transient response that have the high frequency components. The frequency response of the opamp stage including the Rs is reasonably flat until 15 kHz, which is well beyond my area of interest. In addition, the RC constant at the input of the OPAMP is ~3 ns that gives sufficient charge/discharge latency for my input signal.

To post reply to a comment, click on the 'reply' button attached to each comment. To post a new comment (not a reply to a comment) check out the 'Write a Comment' tab at the top of the comments.